This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3. This manual provides documentation for the following instruments: PSA Series. EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). Agilent Technologies. PSA Series Spectrum Analyzer. This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3.
|Published (Last):||16 February 2007|
|PDF File Size:||7.64 Mb|
|ePub File Size:||1.64 Mb|
|Price:||Free* [*Free Regsitration Required]|
User Preset This type of preset restores the currently selected mode to a known Mode Preset Power Requirements The only physical installation of your Agilent spectrum analyzer is a connection to a power source. Page 30 Front and Rear Panel Features Key Overview In other key menus, one key label will always be highlighted to agileng which key has been selected but the menu is immediately exited when a selection is made.
Increments or decrements the current value. For more information on making measurements, see the measurement guide for your analyzer.
e44440a The analyzer was shipped from the factory with the Alignments mode set to Auto. The following six models are available: Viewing Traces 1, 2, and 3 Step 2. Place a marker labeled 1 on the 10 MHz peak, as agjlent in Press Peak Search Note that the frequency and amplitude of the marker appear both in the active function block, and in the upper-right corner of the screen.
Viewing A Signal Viewing a Signal 1. A pound uncoupled function. Page 73 Perform an Alignment. Check to see if the green LED above the power switch is on. Arrow Keys Numeric Keypad Enters a specific value. Options see page Unless specified otherwise, all options are available when you order a spectrum analyzer; some options are also available as kits that you can order and install after you receive the analyzer.
Then press the desired terminator either a unit softkey, or the Terminate a value that requires a unit-of-measurement. If so, refer to the Instrument Messages and Functional Tests manual. Additional Information Notice The information contained in this document is subject to change without notice. Front-panel keys that are white perform an immediate action rather than bringing up a menu.
Accessories Sales and Service Office and are listed below. The file cannot be placed inside another file. Each softkey More has several characters on it.
Page 70 In Case of Difficulty Before calling Agilent Technologies, however or returning an analyzer for servicemznual the quick checks listed in the problem. By pressing ‘print’ button you will print only current page.
AGILENT TECHNOLOGIES EA GETTING STARTED MANUAL Pdf Download.
WARNING If this product is agklent be energized via an external auto transformer for voltage reduction, make sure that its common terminal is connected to a neutral earthed pole of the power supply. If packing foam is not available, the best alternative is S. Front and Rear Panel Features This chapter gives you an overview of the front and rear panels of your analyzer.
File Menu Functions This chapter describes how to use the functions located under the front-panel Data storage and retrieval are handled similarly to that of personal computers PCs.
Making a Basic Measurement This chapter is divided into the following sections: These features are only available when special options are installed. If the printer is ready and the printer setup was successful, a printout of the analyzer display will be printed.
Reset the analyzer by pressing 2. The mains plug shall be inserted only in a socket outlet provided with a protective earth contact. Firmware Revision To view the firmware revision of your analyzer, press call Agilent Technologies regarding your analyzer, it is helpful to have this revision and the analyzer serial number available.
Viewing Catalogs and Saving Files The analyzer stores and retrieves data similarly to the way that a personal computer PC does: Surround the equipment with three to four inches of packing material and prevent the equipment from moving in the carton. It is effective over a frequency range of dc to 2 GHz. To access the Load menu, press 3. Page 53 NOTE You should notice that the trace you saved, in this example trace 3, is loaded to trace 2.
E4440A PSA Spectrum Analyzer, 3 Hz to 26.5 GHz
The numeric keypad is also available for the filename. Installation and Setup Running Internal Alignments Running Internal Alignments Each time mamual analyzer is powered on, the internal alignment routine runs automatically. Viewing a Signal 1. This function can be auto-coupled. Directories are displayed as DIR and drives have nothing listed under Type:.